Institut Teknologi Sepuluh Nopember Surabaya
Kampus ITS Sukolilo - Surabaya 60111
031-5921733 , 5923623
Support (Customer Service) :
Have a problem with your access?|
Please, contact our technical support below:
Dewi Eka Agustina
Ansi Aflacha Putri
Tondo Indra Nyata
Aprillia Tri Wulansari
Moh. Fandika Aqsa
! ATTENTION !
To facilitate the activation process, please fill out the member application form correctly and completely
Registration activation of our members will process up to max 24 hours (confirm by email). Please wait patiently
Please read our User Guide
ITS » Master Theses » Geoteknik S2
Posted by eny at 13/01/2007 13:49:27 • 27871 Views
PENENTUAN KOEFISIEN TEKANAN TANAH KESAMPING "AT REST" (KO) DARI TANAH GAMBUT BERSERAT
DETERMINATION OF AT-REST LATERAL PRESSURE COEFFICIENT (Ko) OF FIBROUS PEAT SOILCreated by :
Wismantara, I Gusti Ngurah Nyoman
|Subject: ||Tanah (Teknologi)|
|Keyword: ||At rest lateral |
Pressure coefficient (Ko)
Fibrous peat soil
One of the most important soil parameters to design bearing capacity of a deep foundation is at rest lateral pressure coefficient (Ko). Ko is the coefficient of lateral earth pressure under condition of no lateral deformation. For sand and clay soils, the Ko value can be obtained using mathematics formula; it is a function of an effective internal friction angle ((ÃÂ¯ÃÂÃÂ¦' ) or plastic index (PI) of the soil. For peat soil, however the Ko value can not be determined using the mathematics formula developed for sand and clay soils mention above. It has to be measured directly from laboratory test using a Ko-test tube. Unfortunately, that apparatus is not available in the market.
Due to that reason , the first step of the research carried out herein is to build the Ko-test tube apparatus; afterward, it is used to determine the Ko value of peat taken from Pekantua, Riau.
The Ko-test tube was built as suggested by Abdelhamid and Krizek (1976); it was calibrated with water to determine the relationship between horizontal strain of the Ko-test tube wall and radial stress. By using the apparatus, the Ko of sand was determined; the result obtained is then compared to the Ko obtained from Jaky's formula. Whenever that apparatus gave a satisfactory result of Ko, the testing was carried on for peat soil. In this research, the vertical load chosen for the test was 50 KPa, 100 KPa, 200 KPa and 400 KPa for loading and unloading conditions.
The results show that the calibration result of the Ko-test tube can be expressed as linear equation, that is, ÃÂÃâr = 2.98 x 106 e average (ÃÂÃâr in KPa). The Ko of sand determined using the Ko-test built is comparable to that obtained using Jaky's formula, that is 0.46. Under loading condition, the Ko value of peat is constant 0.32; under unloading condition the Ko is not constant but it is function of OCR in which the increase of OCR is companied with the increase of Ko; the Ko is variase between 0.32 - 0.44. The relationship between the Ko and OCR can be expressed as a parabolic equation: Ko = 0,4757 OCR0,1579 (for sand) and Ko = 0,321 OCR6,1494 (for fibrous peat from Riau).
- Ir. Noor Endah, MSc., Ph.D.
|Date Create||:||20/03/1998 |
|Format||:||pdf ; 71 pages|
|Call Number||:||624.151 36 Wis|
Theses Civil Engineering RT 624.151 36 Wis p, 1998
Copyright @1998 by ITS Library. This publication is protected by copyright and permission should be obtained from the ITS Library prior to any prohibited reproduction, storage in a retrievel system, or transmission in any form or by any means, electronic, mechanical, photocopying, recording, or likewise. For information regarding permission(s), write to ITS Library
Publication URL :
[ Free Download - Free for All ]
[ FullText Content - Please, register first ]
1. ITS-Master-1344-3195201002-its-judul.pdf - 245 KB
2. ITS-Master-1344-3195201002-its-bab1.pdf - 88 KB
3. ITS-Master-1344-3195201002-its-bab2.pdf - 493 KB
4. ITS-Master-1344-3195201002-its-bab3.pdf - 440 KB
5. ITS-Master-1344-3195201002-its-bab4.pdf - 1158 KB
6. ITS-Master-1344-3195201002-its-babkesimp.pdf - 611 KB
7. ITS-Master-1344-3195201002-its-lampiran.pdf - 602 KB
Give your Comment here...
10 Similar Document...
10 Related Document...
POLLINGBagaimana pendapat Anda tentang layanan repository kami ?
Visitors Today : 1
Total Visitor : 2515760
Hits Today : 41859
Total Hits : 51879744
Visitors Online: 1
16 May 2012
You are connected from 10.199.13.5
using CCBot/2.0 (http://commoncrawl.org/faq/)